Customers Also Bought
Who Uses This Product: Individuals such as diabetics who regularly need to prick their fingers for blood sample testing.
Why They Love This Product: It is compact and offers adjustable settings to find the easiest, most comfortable poke.
Accu-Chek Softclix Lancet Overview
The Accu-Chek Softclix Lancet Device makes capillary blood collection from the fingertips virtually pain-free. Precision-guided technology minimizes any side-to-side motion that may tear skin and cause pain. Thin-gauge, bevel cut lancets provide an effortless entry.
Features and Benefits
- Comfort Dial Depth Adjustment
- Eleven Depth Settings
- Primer and Release Button
- Easy Removal of Used Lancet
Frequently Asked Questions (FAQs)
Can a lancet be reused?
The Centers for Disease Control (CDC) strongly recommends against reusing any lancet. It can lead to bacterial contamination and infection. In addition, reusing a lancet will blunt the tip and cause pain.
Can any kind of lancet be used with the Accu-Chek Softclix?
No. Only Accu-Chek Softclix Lancets are compatible with this lancing device.
Do Accu-Chek Softclix lancets expire?
Yes. The lancets come with a guarantee of sterility for four years.
Does this use a drum or a single lancet?
A single lancet.
Can this be used for non-fingertip testing?
Yes, but it will require an alternate site testing (AST) cap. Purchase additional AST caps separately for the Accu-Chek Softclix Lancets device.
- Manufacturer: Roche
- Brand: Accu-Chek
- Weight: 1.6 Ounces
- Penetration Depth: .8 to 2.33 Millimeters
- Lancet: Beveled, 28-Gauge
- Intended Use: Collection of Capillary Blood
- Warranty: 1 Year
How To Clean and Disinfect the Lancing Device
Note: Always remove the lancet before cleaning or disinfecting. Do not throw away the cap after each use.
Clean and disinfect the lancing device’s surface at least once a week to remove dirt or other material. The Accu-Chek exterior may slightly discolor after multiple cleaning or disinfecting cycles. The discoloring should not affect the lancing device’s functionality. Super Sani-Cloth is excellent for cleaning and disinfecting. Use one cloth for cleaning and a separate cloth for disinfecting.
- Wash hands thoroughly with soap and water.
- Wipe the entire lancing device surface and the inside of the cap.
- With a second Sani-Cloth, disinfect the lancing device in the same fashion as step 2.
Manuals and Documents
- User’s Manual contains instructions for proper use of this lancet device.
How To Use The Accu-Chek Softclix Lancing Device 3:03
- Additional Information