KCI Jobst System 7500 Intermittent Compression Pump provides gradient sequential three chamber compression to manage and reduce peripheral edema with an adjustable pressure range of 1-130mmHg. The gradient sequential compression fills each chamber in sequence up the limb with each subsequent chamber at less pressure than the previous. KCI Jobst System 7500 Compression Pump is compact, lightweight and portable and designed for home use. The KCI Jobst System 7500 has an inflation time of 60 seconds for each chamber and all segments deflate simultaneously in 50 seconds.
The KCI Jobst System 7500 Intermittent Compression Pump can operate two garments at once but requires a separate adapter see item # NTI147520a_ea (Adapter to operate two garments at once). The KCI Jobst System 7500 Intermittent Compression Pump has a long history for treating lymphedema and for enhancing lymphatic and venous flow. The KCI Jobst System 7500 Intermittent Compression Pump comes with a choice of three garment sizes to treat a variety of vascular and lymphatic conditions.
Jobst Compression Sleeves Measurement Guidelines:
For arm length measurement: Measure axilla (arm pit) to finger tip.
For leg length measurement: Measure inseam (inner thigh) to ankle.
For arm circumference: Measure at axilla (arm pit).
For leg circumference: Measure thigh 2" below groin.
For ankle circumference: Measure 1" above talus (ankle bone).
*Note when selecting sleeve length the total sleeve length should never encroach within 1 ½ inches of your groin or underarm or circulation can be impeded.
KCI Jobst 7500 Comparison ChartKCI Jobst System 7500
Sequential Compression PumpKCI Jobst System 7000
Intermittent Compression Pump
Product CodeNTI147500NTI147000 Power Supply110-120V AC 60Hz110-120 V, 60 Hz, 15 Watts Size
(WxHxL)8.75 x 4.5 x 10.75"8.75 x 4 x 10.75"
Electrical110/120 V, 60 Hz, 15 Watts110/120 V, 60 Hz, 15 Watts Pressure
Range1-100 mmHg (user adjustable)1-100 mmHg (user adjustable)
Cycle TimeDistal SegmentMiddle SegmentProximal Segment Seconds to inflate606060 Seconds at setting1307010 Total Inflate19013070 Seconds to deflate
90 seconds inflation
30 seconds deflation
This product is designed to aid in the reduction and control of peripheral edema, including congenital lymphedema of the lower extremities and postmastectomy lymphedema, stasis dermatitis and venous stasis ulcers.
This product should not be used during the inflammatory phlebitis process or during episodes of pulmonary embolism, congestive heart failure, pulmonary edema, suspected deep vein thrombosis or in any instances where increased venous and lymphatic return is undesirable. Use with caution on an extremity which is not sensitive to pain.
KCI Jobst 7500 Compression Pump Features & Benefits
- Sequentially inflates air chambers distally to proximally
- Ability to treat two extremities simultaneously (with adapter tubing)
- Color-coded pressure gauge
- Adjustable pressure control
- Pre-set inflation and deflation cycle
- Safety alarm system
- Compact, lightweight and portable make it ideal for home use
- U.L. listed for home use
- Kink-resistant, color coded tubing
Compression Sleeves Features & Benefits
- Color-coded ports simplify set-up
- Anti-reflux safety valves (designed to help prevent reverse inflation)
- Soft nylon pile inner lining
- Durable construction
- Machine washable
- Longitudinal zipper for ease of application
- Variety of sizes available
- Additional Information
Top Rated Customer Reviews - See All Reviews
- GREAT CUSTOMER SERVICE !!
The pump is no longer available even though it is online. Customer Service did an EXCELLENT job in helping me find an equivalent pump from another manufacturer and even though the other manufacturer price was higher they charged me the lower price. This company has GREAT customer service and the person I spoke with was Kaylene and she was very apologetic, pleasant and concerned about what happened. Good customer service is hard to find but Vitality Medical customer service should be recommended because they care about the customer. I will definitely buy from them again because Kaylene in customer service handled this situation so well.
- Works great
The machine works very well; appears to be the equivalent to a much more expensive machine which I used in Canada. I'm happy.
- Community Q&A
Product QuestionsNo questions yet. Be the first to ask the question!