Aeroneb Go Micropump Nebulizer is a Nebulizer Compressor that utilizes micropump technology to operate silently while delivering respiratory medications rapidly with a minimum of waste. The Aeroneb Go Nebulizer, manufactured by Respironics, formerly by Drive Medical, uses passive delivery to avoid the pooling of medication in back of the throat. The Aeroneb Go ensures proper delivery of medication to the lungs where it is needed. The handset is very lightweight at only 2 ounces and is quick and easy to use. Operation of the Aeroneb Go Nebulizer is virtually silent. The aperture plate used on the Aeroneb Go Micropump Nebulizer has approximately 1,000 precision formed tapered holes to produce a uniform mist.
Aeroneb Go Micropump Nebulizer Features & Benefits
- Near silent operation so that you hardly know it is running.
- Lightweight handset is easy to use.
- Delivers standard 3 cc dose in less than 5 minutes.
- Vibrating mesh technology.
Aeroneb Go Micropump Nebulizer Specifications
- Width: 4.1”
- Length: 1.6”
- Height: 3.7”
- Handset Weight: 2 Ounces.
- 2 Selection Options: Battery Powered, 7091; and Combo Battery & AC Powered, 7090.
- Flow: >0.25 mL per minute.
- Battery: 4-5 VDC, 350 mA, 3 AA disposable alkaline or lithium batteries (not included).
- HCPCS code: E0574 and E0570
- Aeroneb Go Micropump Nebulizer Machine: 1 year.
- Aeroneb Go Micropump Nebulizer Battery Controller: 2 years.
- Aeroneb Go Micropump Nebulizer AC Controller: 2 years.
Aeroneb Go Instructional (1:48 Minutes)
In this video, Evo Medical Solutions is going to give you a step by step process, walking you through the assembly and functions of the Aeroneb Go Micropump Nebulizer. The Aeroneb Go is a very quick and efficient treatment method, with hassle free operation and a lightweight handset. It can be operated by both battery and AC power. Before beginning, make sure you run the medication cup under hot water or dip it in and out of boiling water. First, connect the medication cup to the nebulizer body, making sure it is unplugged before adding the medication. Fill the medication cup with the prescribed medication and never pour more than 6cc’s into the cup. Once, the medication cup is filled, securely place the cup on top of the medication cup. Then attach the mouth piece to the nebulizer body. Masks can also be attached in this way. Once, the nebulizer has been assembled, you have the option to either insert the batteries, or attach the AC cable to the remote which plugs directly into the wall. On battery power, the remote will automatically shut off after 15 minutes. Next, connect the nebulizer cable to the powered remote and then into the back of the nebulizer unit. Finally, turn on the nebulizer by pressing the ON-OFF button, located at the top of the remote. You should visually be able to see the nebulizer producing aerosol medication. The Aeroneb Go is able to aerosolized a wide variety of medications right down to the last drop. Treatments can be achieved in under 5 minutes every time.
Aeroneb Go Micropump Nebulizer Video
- Additional Information
- Community Q&A
Product QuestionsNo questions yet. Be the first to ask the question!