Philips Respironics CPAP Battery Pack is designed to provide uninterruptible power to your CPAP device even during power outages. Manufactured by Philips Respironics, the battery is lithium ion with a long charge life. This CPAP Battery Kit is also known as the PAP Battery Kit. This versatile charger plugs into an AC power source to recharge the external battery. The battery pack can power your CPAP device for more than 14 hours before needing a recharge.
The PAP Battery Kit works with the following PAP devices: DreamStation and System One 50/60 series CPAP, BiPAP auto, BiPAP autoSV, BiPAP AVAPs, BiPAP ST, REMstar SE, and the REMstar Plus. Replacement Parts for the CPAP Battery Pack are also available.
Philips Respironics also makes a battery pack kit for respiratory care that powers the Trilogy.
Respironics CPAP Battery Kit Components
- Lithium ion battery pack
- LithiumIon battery, PAP device cable
- 50 series Cable Adapter, (#2)
- Battery Case
PAP Battery Kit Features & Benefits
- Power source when power outage occurs.
- Provides convenience when traveling.
- 90 Watt battery pack.
CPAP Battery Kit Specifications
- Product Number: 1124009.
- Dimensions: 1.25 x 5.25 x 9 Inch.
- CPAP Mode
- Pressure 10 cm H2O
- No heated humidification or heated tubing
- Weight: 2.25 lbs.
- Power Capacity: 90 W.
- Output: 11 to 15 VDC.
- Battery Type: Lithium Ion.
- Application CPAP Uninterruptible Power Supply.
- Run Time: PAP use of up to 14 hours.
- Recharge Time: 3 to 4 hours.
- Minimum Life Cycle: 70% or more of rated capacity after 1000 cycles.
- Precription: Not required.
- Manufacturer: Philips Respironics.
- Respironics CPAP Battery Kit: Warranty: 6 Months
Philips Respironics PAP Battery Kit Review (4:31 Minutes)
The lithium ion battery pack is an accessory for use with certain Philips respironics sleep therapy and non-invasive ventilation devices. It comes with a carrying case, battery pack, cable, adaptors, and instructions for use. To charge the pack, you will also need the power supply that came with your therapy device. Before first use, the pack will need to be fully charged, which may take up to three hours. Plug one end of the power supply into an AC outlet and the other into the battery pack. It will begin to charge automatically. The LED display will light up momentarily to show the charge level. To check the charge level, press the push button on the LED display. A blinking first light indicates a zero to 20% charge, the second, 21-40%, the third, 41-60%, the fourth 61-80%, and the fifth 81-99%. The battery pack is fully charged and ready for first use when you press the push button and all LEDs are steadily lit. If you are charging your battery after it is completely depleted, it may take a few minutes for the LEDs to light up. During operation, your battery pack can be used in uninterruptible power mode or in standalone mode. Uninterruptible or UPS mode allows you to use the battery pack continuously without losing any charge and allows your device to operate during a power outage. To use the battery pack in uninterruptible mode, plug one end of the power supply into an AC outlet. Connect the other end into the battery pack and wait for the LEDs to stop blinking. In order for your therapy device to detect what power is available, it is important to connect the components in the proper sequence. Insert the strap attached to the cable through the slot in the battery pack. Connect the therapy device cable to the battery pack. Pull the strap through. Wrap the strap around the therapy device cable and secure it in place. Connect the other end of the therapy device cable into the device. Power will automatically be provided. Some therapy devices require an adaptor. To check if yours does flip the device over and if the part number includes a 50 you need to use an adaptor with the device cable. Set up for standalone mode is similar except that the battery pack is not connected to AC power. Check to make sure that the battery pack is charged by pressing on the push button located on the LED display. As the battery discharges, fewer LEDs illuminate. Five lights indicate a charge of 81-100%; four- 61-80%; three-41-60%; two- 21 to 40%; and one - 11 to 20%. When there is one blinking light, there is only one to ten percent charge remaining in the battery. Once the battery is charged, thread the cable strap through the slot. Connect the therapy device cable into the battery pack and secure it in place. Connect the loose end of the cable to your device adding an adaptor if needed. The battery pack will then power the sleep therapy device. To disconnect the battery pack simply unplug the therapy cable. The battery pack can be transported in the carrying case. Use the battery pack in UPS mode to keep it fully charged and ready for travel. Innovation and you. Philips Respironics.
- Additional Information
Manufacturer Respironics Ships Free Yes
- Community Q&A
Product QuestionsNo questions yet. Be the first to ask the question!